Breast imaging – Mammography technique – Pitfalls in techniques of mammography – Screen artefacts on mammographic image
Interference with image viewing, especially evaluation of breast
Probable cause – screen artefacts on mammographic image
- When the latent image is processed in the computed radiography system, the screen is passed through rollers and photomultipliers to convert it to a visible image. Abrasion of the screen in the computed radiography system rollers may cause linear screen artefacts.
Optimum maintenance and quality check of the CR system performance allows smooth movement of the rollers and post processing of the latent image to obtain image of adequate diagnostic quality.
- Ensure that the computed radiography system is regularly maintained and checked for quality. The rollers should move smoothly, and post-processing of the latent image should produce an image of adequate diagnostic quality.
Click on the pictures to magnify and display the legends
Click on this icon to display a case study
IARC, 150 Cours Albert Thomas, 69372 Lyon CEDEX 08, France - Tel: +33 (0)4 72 73 84 85 - Fax: +33 (0)4 72 73 85 75
© IARC 2023 - All Rights Reserved